The NHancer Pro X is a single use device, consisting of a hub with strain relief, shaft, distal tip and torquer. The strain relief is to avoid kinking of the shaft. On the shaft depth markings are applied to indicate that the distal tip of the catheter is at the level of the distal tip of the guiding catheter (95 and 105 cm). As aid in positioning the tip of the catheter the distal tip is provided with radiopaque material to make the distal tip visible with fluoroscopy. The device provides a torquer that can be used to lock the proximal part of the catheter to the wire in such way that the guide wire and the NHancer Pro X act as being one integral device.
- Re-enforced shaft for optimal push
- Torquer for optimal maneuverability
The catheter is intended to support a guide wire during access of vasculature and allows for exchange of guide wires and provide a conduit for the delivery of diagnostic contrast agents.
A guide wire is used to bring interventional devices (PTCA balloon catheters; Stent delivery systems) to and into the occlusion.
Procedures with a high level of complexity require enhanced guide wire performance which is not always possible to achieve with conventional devices.
The purpose of the NHancer Pro X is to facilitate the placement of a guide wire through an occluded vessel during Percutaneous Transluminal Coronary Intervention procedures. This by protecting the wire and giving the wire stability.
|Technical Specification||NX31413525 (A-NX3)||NX31415560 (R-NX3)||NX61413525 (A-NX6)||NX61415560 (R-NX6)|
|Guide wire size||0.014″||0.014″||0.014″||0.014″|
|Usable length||135 cm||155 cm||135 cm||155 cm|
|Hydrophilic coating length||25 cm||60 cm||25 cm||60 cm|
|Proximal shaft outer diameter||3.0 F||3.0 F||2.6 F||2.6 F|
|Distal shaft outer diameter||2.0 F||2.0 F||2.3 F||2.3 F|
|Proximal shaft inner diameter||0.52 mm||0.52 mm||0.52 mm||0.52 mm|
|Distal shaft inner diameter||0.45 mm||0.45 mm||0.48 mm||0.48 mm|